|High-speed air turbine handpiece||Idling speed ≥ 35 x 10 r/min (Air pressure:0.22 Mpa / 32 PSI)|
|Low-speed air motor handpiece||Idling speed ≥ 18 x 10³ r/min (Air pressure:0.30 Mpa / 44 PSI)|
|Dental light maximum illuminance||≥ 25000 lux|
|Electrical patient chair carrying capacity||≥ 1323 N (= 135KG) / ≥ 298 lb|
|The higest position of the patient chair from the ground||≥ 780 mm / ≥ 31 lnches|
|The lowest position of the patient chair from the ground||≥ 380 mm / ≥ 15 lnches|
|Backrest movement range||110° ~ 180°|
|Head rack extension range||0 ~ 100 mm / ≥ 4 lnches|
|Dimensions (length* width* height)||1950 mm x (1200 mm ~ 1800 mm) x 2000 mm|
|Input power supply||300 VA|
|Air supply pressure||0.55 ~ 0.80 Mpa / 80 ~ 116 PSI|
|Water supply pressure||0.20 ~ 0.40 Mpa / 29 ~ 58 PSI|
|Power cable requirement||3 x 0.75 mm² , according to UL 60601-1 CAN/CSA C22.2, No.601.1|
|Electrical classification||Group 1|
|EMC classification||Group 1, Class B|
|Vacuum||12 cfm (339.84 L/min) @ 8 inches of mercury|
"I have used other well known brands of equipment in my office and have noticed that your equipment is far less problematic than others. It has proven that my decision to install Beyes equipment was the right one! I also saved considerable money compared to the traditional name brands of equipment. Service response time is impressive, any questions or concerns have been taken care of very quickly and efficiently. "
Dr. Safwat Khalil
"I have been using the Comfortek Chair from Beyes Dental for over a year now. I am quite impressed by the simplicity of this unit. It is reliable and easy to operate . The chair has a variety of features ; specifically, a built-in intraoral camera and pizzo that facilitate it's operation with ease. The chairs are also cost effective. I would recommend these chairs without hesitations."
"I am a general dentist working in a solo practice. Having been in this office for almost 20 years, it was the right time to replace my dental chair. My previous experience with both Siemens(Sirona) and Adec has been relatively positive; therefore I was looking to buy the same brand. However, I was surprised to see that a dental chair was available at Beyes where I have been purchasing my dental supplies. With the opportunity to see and feel the chair at their location, it certainly looked remarkable and felt comfortable to use. I was still hesitant to purchase because the price was exceedingly below my previous budget of a new dental unit. I was reassured , however, because my experience with Beyes has been quite positive. Many other products that I have purchased from them have been very reasonably priced and functions well such as their curing lights and ultrasonice scaling units. I took the plunge and bought one to try it out. It has been over one month since the purchase. Overall I am very happy with the purchase. The chair has better functionality than my previous units and their service has been exceptional. Patients have remarked how 'updated and high tech" this chair looks and I have no problem adjusting to using this chair. The only regret is that I didn't replace both dental units at once to complete the transition of updating my office. "
Dr. Ivan Mo
"We recently purchased the Comfortek Dental Chair for a dentaly hygiene operatory. We find it meets all our expectations for both staff and patient efficiency and comfort.Thank you for your excellent customer service."
Samuel Hait, DDS